|Re: Re: True Position with MMC|
|Re: Re: True Position with MMC -- John-Paul Belanger||Post Reply||Top of thread||Forum|
Posted by: Antonio Patino |
Just like John said, be very careful, CMM does not always consider MMC for the datum unless you add a rutine to calculate it and add the extra tol. I once had a situation in which one piece did OK on the inspection fixture but rejected on CMM, the problem was that CMM was not considering the extra tol given by the MMC of the datums.
|Post Reply | Recommend | Alert||Rate||Where am I? Original Top of thread|
Powered by Engineers Edge
© Copyright 2000 - 2019, by Engineers Edge, LLC All rights reserved. Disclaimer